## 1/f Noise Analysis for FAST HI Intensity Mapping Drift-Scan Experiment

### AI-generated Key Points

- The study investigates the 1/f noise of the Five-hundred-meter Aperture Spherical Telescope (FAST) receiver system using drift-scan data from an intensity mapping pilot survey.
- The FAST has a large aperture and is equipped with multiple feeds and receivers, making it ideal for conducting large surveys.
- The observations were carried out with the L-band 19-beam feed system and the associated cryogenic receiver.
- All 19 beams have 1/f fluctuations with similar structures, which can be important for HI intensity mapping.
- Data was collected by the FAST Hi intensity mapping pilot experiment during risk-shared observations made in May 2019.
- Both temporal and two-dimensional power spectrum densities were estimated, showing that correlations directly seen in the time series data at low frequency $f$ are associated with sky signal, likely due to a coupling between foregrounds and system response.
- To remove foregrounds, Singular Value Decomposition (SVD) was used to subtract the strongest components of measured noise power significantly reducing it by up to twenty modes subtraction; thus reducing knee frequency of noise in a ten MHz band to $1.8 \times10^{-3}\Hz$, well below thermal noise over a scale of five hundred seconds.
- A simple noise diode frequency-independent calibration of receiver gain at eight-second intervals did not affect results as the 1/f noise had a knee frequency of $(f_{k}) \sim$ 6 $\times$ 10$^{-4}$Hz and time and frequency correlation spectral indices $(\alpha) \sim 0.65$, $(\beta) \sim 0.8$ after SVD subtraction of thirty modes respectively .
- Residual 1/f noise can bias HI power spectrum measurement by up to ten percent; making it important to consider in HI intensity mapping experiments conducted using FAST receiver systems.
- The study provides valuable insights into characterization and mitigation of 1/f noise crucial for current optical surveys.

**Authors:**
Wenkai Hu,
Yichao Li,
Yougang Wang,
Fengquan Wu,
Bo Zhang,
Ming Zhu,
Shifan Zuo,
Guilaine Lagache,
Yinzhe Ma,
Mario G. Santos,
Xuelei Chen

**Abstract:** We investigate the 1/f noise of the Five-hundred-meter Aperture Spherical Telescope (FAST) receiver system using drift-scan data from an intensity mapping pilot survey. All the 19 beams have 1/f fluctuations with similar structures. Both the temporal and the 2D power spectrum densities are estimated. The correlations directly seen in the time series data at low frequency $f$ are associated with the sky signal, perhaps due to a coupling between the foreground and the system response. We use Singular Value Decomposition (SVD) to subtract the foreground. By removing the strongest components, the measured 1/f noise power can be reduced significantly. With 20 modes subtraction, the knee frequency of the 1/f noise in a 10 MHz band is reduced to $1.8 \times 10^{-3}\Hz$, well below the thermal noise over 500-seconds time scale. The 2D power spectra show that the 1/f-type variations are restricted to a small region in the time-frequency space and the correlations in frequency can be suppressed with SVD modes subtraction. The residual 1/f noise after the SVD mode subtraction is uncorrelated in frequency, and a simple noise diode frequency-independent calibration of the receiver gain at 8s interval does not affect the results. The 1/f noise can be important for HI intensity mapping, we estimate that the 1/f noise has a knee frequency $(f_{k}) \sim$ 6 $\times$ 10$^{-4}$Hz, and time and frequency correlation spectral indices $(\alpha) \sim 0.65$, $(\beta) \sim 0.8$ after the SVD subtraction of 30 modes. This can bias the HI power spectrum measurement by 10 percent.

### Ask questions about this paper to our AI assistant

You can also chat with multiple papers at once here.

Assess the quality of the AI-generated content by voting

Score: 0

**Why do we need votes?**

Votes are used to determine whether we need to re-run our summarizing tools. If the count reaches -10, our tools can be restarted.

## Similar papers summarized with our AI tools

Navigate through even more similar papers through a

tree representation**Look for similar papers (in beta version)**

By clicking on the button above, our algorithm will scan all papers in our database to find the closest based on the contents of the full papers and not just on metadata. Please note that it only works for papers that we have generated summaries for and you can rerun it from time to time to get a more accurate result while our database grows.

**Disclaimer:** The AI-based summarization tool and virtual assistant provided on this website may not always provide accurate and complete summaries or responses. We encourage you to carefully review and evaluate the generated content to ensure its quality and relevance to your needs.