Measuring a Charge-Coupled Device Point Spread Function: Euclid Visible Instrument CCD273-84 PSF Performance
AI-generated Key Points
⚠The license of the paper does not allow us to build upon its content and the key points are generated using the paper metadata rather than the full article.
- The authors present a novel modelling technique for measuring the intrinsic Charge-Coupled Device (CCD) Point Spread Function (PSF) characteristics of a back-illuminated development Euclid Visible Instrument (VIS).
- Optical spot projection system and the CCD273-84 PSF are jointly modelled using Bayesian posterior probability density function sampling to all available data simultaneously.
- Characterisation of a CCD273-84 PSF as a function of wavelength and intensity reveals that the CCD PSF kernel size increases with increasing intensity and decreasing wavelength.
- This work has important implications for understanding the performance of CCDs in astronomical observations, particularly relevant for future space missions such as Euclid.
- Detailed simulations are included to demonstrate their approach which could be useful for other researchers working in this field.
- Overall, this paper presents an innovative method for characterising CCD PSFs that has broad applications in astronomy and beyond.
Authors: Sami-Matias Niemi, Mark Cropper, Magdalena Szafraniec, Thomas Kitching
Abstract: In this paper we present the testing of a back-illuminated development Euclid Visible Instrument (VIS) Charge-Coupled Device (CCD) to measure the intrinsic CCD Point Spread Function (PSF) characteristics using a novel modelling technique. We model the optical spot projection system and the CCD273-84 PSF jointly. We fit a model using Bayesian posterior probability density function, sampling to all available data simultaneously. The generative model fitting is shown, using simulated data, to allow good parameter estimations even when data are not well sampled. Using available spot data we characterise a CCD273-84 PSF as a function of wavelength and intensity. The CCD PSF kernel size was found to increase with increasing intensity and decreasing wavelength.
Ask questions about this paper to our AI assistant
You can also chat with multiple papers at once here.
⚠The license of the paper does not allow us to build upon its content and the AI assistant only knows about the paper metadata rather than the full article.
Welcome to our AI assistant! Here are some important things to keep in mind:
- The assistant will only answer questions related to this specific paper.
- Please note that this is not a bot for casual chatting.
- If you want the answer in a language other than the language you chose for navigating the website, simply add "TRANSLATE IN LANGUAGE L" at the end of your query (replace "LANGUAGE L" with the language of your choice).
- For example, you could ask "Can you extract the most important aspect of the paper? TRANSLATE IN SPANISH".
- If you want to keep the history of your questions/answers you should create an account.
Assess the quality of the AI-generated content by voting
Why do we need votes?
Votes are used to determine whether we need to re-run our summarizing tools. If the count reaches -10, our tools can be restarted.
Similar papers summarized with our AI tools
Navigate through even more similar papers through atree representation
Look for similar papers (in beta version)
By clicking on the button above, our algorithm will scan all papers in our database to find the closest based on the contents of the full papers and not just on metadata. Please note that it only works for papers that we have generated summaries for and you can rerun it from time to time to get a more accurate result while our database grows.
Disclaimer: The AI-based summarization tool and virtual assistant provided on this website may not always provide accurate and complete summaries or responses. We encourage you to carefully review and evaluate the generated content to ensure its quality and relevance to your needs.